INSTALLATION / OPERATIONAL QUALIFICATION PROTOCOLThe following Installation Qualification (IQ) & Operation Qualification (OQ) isintended to be a guideline. Local IQ & OQ procedures can vary depending onintended use. The IQ & OQ is concluded if all the criteria of acceptance are verifiedand the possible deviations are accepted.This IQ & OQ is intended for the following products:TabletopULTF 40UndercounterVTS 098UprightVTS 252VTS 254VTS 256VTS 258LocationModelSerial numberName of distributorWarranty StartSignatureDateDocument Reference: IOQ- ManualsRevision: 04-05-2020Rev. 04-05-20201/6
Acceptance criteriaThe IQ/OQ is concluded if all criteria of acceptance are approved and the possible deviations are accepted.ReferenceUser manualPersons performing IQ & OQAll personnel entering information in this protocol must be identified in the table below by name, fullsignature and initials.Personnel must be fully aware of all the procedures to follow.NameCompanySignatureInitialsDate2/6
IQ - Installation QualificationObjectiveTo verify that the freezer has been installed to a satisfactory standard and is safe to use.Pre-RequisitesUser manual is available for reviewStepIQ-1Reference tochapter inmanualBefore UseIQ-2Before UseIQ-3IQ-4Before UseBefore UseIQ-5Before y backupLocation andventilationLocation andventilationLocation 4Location andventilationLocation andventilationLocation andventilationDescriptionVerificationPass FailCommentsEnsure the appliance has not been damaged duringtransport.Ensure the protective foil on the shelves areremoved before cleaning and using the unit.Ensure that the inside of the appliance is clean.Ensure the appliance has been in an uprightposition for at least an hour before being switchedon.Allow the freezer to operate at the desiredtemperature for a minimum of 12 hours beforeloading.Ensure to connect the appliance to the socketoutlet.Ensure the ventilation guards are mounted behindthe appliance.Ensure the battery backup system is activated.Ensure the appliance is not exposed to directsunlight.Ensure the appliance is not exposed to rain.Ensure the appliance is installed in a dry, wellventilated room with minimal variations intemperature.Ensure the appliance is away from sources of heat.Ensure ventilation, above, below and around theappliance.Ensure enough space for the door to open in atleast 90 degrees.3/6
IQ-15Levelling theapplianceEnsure the appliance is levelled.ConclusionThis section of the protocol has been completed and is deemed to (initial appropriate box below):PassPass with commentCommentsFailNon-conformance4/6
OQ - Operation QualificationObjectiveTo verify that all refrigerator components function correctly as specified for normal operation.Pre-RequisitesAppliance is ence tochapter inmanualFreezerComponentsController –operation andfunctionController –operation andfunctionController –operation andfunctionController –operation andfunctionController –operation andfunctionController –operation andfunctionController –operation andfunctionDescriptionVerificationPass FailCommentsEnsure all doors operate as expected.Ensure the appliance is turned on.Ensure the time and date are set on the controller.Hold the door open to confirm the alarm activates.Close door and confirm alarm de-activates.Record temperature set points (for both operatinglevel and the alarm limits) and confirm these meetthe acceptance criteria.Increase the temperature set point by 11oC (forULT units its by 26oC). Confirm that therefrigerator system responds to increasedtemperature and the alarm activates (visual andaudible) when temperature exceeds alarm highlimit.Ensure temperature set point is returned to theoriginal value on completion and confirm alarm deactivates.Decrease the temperature set point by 11oC.Confirm that the refrigerator system responds todecreased temperature and the alarm activates(visual and audible) when temperature exceedsalarm low limit.5/6
OQ-9Controller –operation andfunctionEnsure temperature set point is returned to theoriginal value on completion and confirm alarm deactivates.ConclusionThis section of the protocol has been completed and is deemed to (initial appropriate box below):PassPass with commentCommentsFailNon-conformance6/6
INSTALLATION / OPERATIONAL QUALIFICATION PROTOCOL Rev. 04-05-2020 1/6 The following Installation Qualification (IQ) & Operation Qualification (OQ) is intended to be a guideline. Local IQ & OQ procedures can vary depending on intended use. The IQ & OQ is concluded if all the criteria of acceptance are verified and the possible deviations are .
The installation qualification must be completed prior to execution of the operational qualification phase. Operational Qualification Operational qualification demonstrates that the installed equipment is functioning within established limits as specified by Helmer. The operational qualification must be completed prior to the execution of the .
OPERATIONAL QUALIFICATION PROTOCOL FOR AUTOMATIC HIGH SPEED LINEAR VIAL WASHING MACHINE-300 Protocol cum Report No.: Batch Size: Page No.: 3 of 19 SCHOLARS QUALITY DEPARTMENT Signing of this Operational Qualification Protocol indicates agreement with the Validation Master Plan approach of the equipment.
Operational qualification Operational qualification (OQ) 13. Operational qualification (OQ) should follow Installation qualification. 3.10 OQ normally follows IQ but depending on the complexity of the equipment it maybe performed as a combined Installation/Operation Qualification (IOQ). OQ could include but is not be limited to the following:
OPERATIONAL QUALIFICATION PROTOCOL FOR AUTOMATIC VISUAL VIAL INSPECTION MACHINE Protocol cum Report No.: Batch Size: Page No.: 4 of 20 QUALITY DEPARTMENT 2.0 OBJECTIVE The objectives of this Operational Qualification (OQ) are as follows: To verify that the equipment operates in accordance with the design and user requirements as .
INSTALLATION QUALI INSTALLATION QUALIFICATION FICATION (1) INSTALLATION INSTALLATION QUALIFICATION QUALIFICATION (1) 1 rue du Petit Clamart - B.P. 50 78142 VELIZY CEDEX - FRANCE Tél. 01.46.29.99.60 - Fax. 01.46.30.87.30 ETUDES & DEVELOPPEMENT VALIDATION C 1/4 NSCA 161/16 LC reference : 37K53E Page Rev. Page Customer reference : 45382135/392 IQ .
5.D Qualification of water supplies 5.D.1 Introduction 5.D.2 Risk analysis 5.D.3 Design qualification 5.D.4 Installation qualification 5.D.5 Operational qualification (OQ) 5.D.6 Transfer to the user 5.D.7 Process validation/performance qualification (PQ) 5.D.8 Qualification report 5.E Operation of water supplies
An equipment qualification is handled in different phases of the equipment lifecycle, start-ing with design qualification, installation qualification, and then operational qualification and performance qualification. Every step in the equipment qualification formally comprises a plan, the execution and a report.
The objective of this protocol is to define the Installation Qualification (IQ) and Operational Qualification (OQ) requirements and acceptance criteria for the [insert system name and plant number] which will be located in the [insert area, packaging or manufacturing] at site [insert site name].
8. Installation Qualification, Operational Qualification and Process Qualification page 30 9. Example of Validation Protocol page 33 10. Example of a qualification Proforma page 35 11. Validation sign off report page 50 12. Acknowledgements and Declarations of interest page 51
The Installation / Operational / Performance Qualification protocol is a comprehensive document, which will be used to guide the executants, in the verification of the proper installation and operation of the Refrigerator located at Xxxxx. 4 OBJECTIVE The purpose of this Installation / Operational / Performance Qualification is to:
The Installation / Operational / Performance Qualification protocol is a comprehensive document, which will be used to guide the executants, in the verification of the proper installation and operation of the Cold Room located at Xxxxxx. 4 OBJECTIVE The purpose of this Installation / Operational / Performance Qualification is to:
The operational qualification serves as proof that the equipment operates as designed and intended as well as fulfills acceptance criteria defined and stated in the Operational Qualification documentation. These criteria are defined and are based on the equipment technical specifications of the manufacturer.
Installation Qualification and Operational Qualification Miris Ultrasonic Processor COMPANY CONFIDENTIAL FORM-943 Installation Qualification and Operational Qualification Miris Ultrasonic Processor Version: 1 Page 1 of 13 Figure 1. The Miris Ultrasonic Processor 1. Sound reducing cabinet 2. Power supply unit
EGP Exterior Gateway Protocol OSPF Open Shortest Path First Protocol IE-IRGP Enhanced Interior Gateway Routing Protocol VRRP Virtual Router Redundancy Protocol PIM-DM Protocol Independent Multicast-Dense Mode PIM-SM Protocol Independent Multicast-Sparse Mode IGRP Interior Gateway Routing Protocol RIPng for IPv6 IPv6 Routing Information Protocol PGM
Operational Qualification section. Labconco Purifier Logic Series BSC’s IQ/OQ/PQ Protocol #1058603 Revision A 3 A. Installation Qualification Step Description Specification or Acceptance Criteria Result YES NO 1 Site Planning 1a Proper airflows Is the Purifier Logic to be located in a room .
covers the three stages of qualification needed to release a temperature-controlled storage area for routine use: installation qualification (IQ), operational qualification (OQ) and performance qualification (PQ). Related topics are covered in the following Technical Supplements: Temperature mapping of storage areas.
Certificate of Operational Qualification This is to certify that I have checked the following verifier today: Axicon verifier model: Serial number: in accordance with the Axicon Operational Qualification test protocol. The verifier will now operate in accordance with Axicon’s specifications and has been configured to meet the user’s .
This installation and operational qualification (IOQ) protocol aims to bring the documented evidence that the equipment described in Table 1 have been installed, configured appropriately, and function as expected, according to company requirements specifications, intended use, and applicable manufacturer's recommendations.
This document package is intended to serve as a guide during the Design Qualification, Installation Qualification, Operational Qualification, Performance Qualification protocol. It outlines how the freezers are to be tested, and documented in order to conform to all of the above requirements and acceptance criteria.
2018 Accounting Higher Finalised Marking Instructions Scottish Qualifications Authority 2018 The information in this publication may be reproduced to support SQA qualifications only on a non-commercial basis. If it is reproduced, SQA should be clearly acknowledged as the source. If it is to be used for any other purpose, written permission must be obtained from [email protected] Where .